This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate, was assessed and found to comply with the IECEx Quality system requirements. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
This Unit verification certificate is issued as verification that the Apparatus identified on page 1, was assessed and tested and found to comply with the IEC Standard list below. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
The electrical apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards:
As this is a Unit Verification Certificate, no QAR is applicable as this certificate is specific to the items listed by serial number or other unique identification.
Description of Equipment
Type 5300 series Guided Wave Radar and 5400 series Radar Level Transmitter, for measurement of liquid level, liquid level interface and solid level transmitter. The transmitters are designed for various signal protocols: Hart, Foundation Fieldbus, FISCO and RS-485. The transmitters are intrinsically safe as a whole, ‘Ex ia’, "Ex ib"or a combination with non-intrinsically safe external signal connection to the ‘Ex d’ terminal compartment.
The ‘Ex d’ compartment is made with two threaded cable entries that need to be fitted with an appropriately certified cable gland and any unused opening need to be closed with a certified closing device. The cable gland and closing device should also have a degree of ingress protection at least IP 66 or IP67
Further description in the ANNEX to this certificate
Special conditions 1, 2, 3, 4, 5 in the ANNEX to this certificate
Type 5400 FoundationTM Fieldbus Models 5401xFxE1 or 5401xFxKG…Ex iad IIC T4 Ta:-40
Issue 1. The changes concern alternative safety components. The changes do not alter the safety characteristics of the product. The changes are included in the test report NO/NEM/ExTR06.0001/01-68314 Standards IEC 60079-0 Ed. 3.1 IEC 60079-11 Ed 4.0 Issue 2. Change of Manufacturers name to Rosemount Tank Radar AB. New variations model 5300-series added. Further description in ANNEX 2.Minor changes of electronics for 5400 series.Additional conditions for this certification:Probes covered with plastic and/or with plastic discs will have a non-conducting area that exceeds the maximum permissible areas for Group IIC and Zone 0 IEC 60079-0. Therefore, when the antenna is used in a potentially explosive atmosphere, appropriate measures must be taken to prevent electrostatic discharge.Standards IEC 60079-0 Ed. 3.1 IEC 60079-1 Ed. 5 IEC 60079-11 Ed. 4.0 IEC 60079-27 Ed. 1 Issue 3. Minor changes of components and boards. Minor mechanical changes. Revision of documents. The changes do not alter the specified safety characteristics of the equipment. Issue 4. Minor changes of electronics and document revisions. New Annex amended and replaces previous Annexes to issue 0 and 2.Issue 5. Change of transformer and PCB, change of coating, alternative components and other changes and revisions of documents.Issue 6. Changes of electronics, added variation of PMMDC 5300 board, document changesIssue 7.Update to new standards. Connection to [Ex ib] power supply, Modbus, Dust protected.