This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate, was assessed and found to comply with the IECEx Quality system requirements. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
This Unit verification certificate is issued as verification that the Apparatus identified on page 1, was assessed and tested and found to comply with the IEC Standard list below. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
The electrical apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards:
As this is a Unit Verification Certificate, no QAR is applicable as this certificate is specific to the items listed by serial number or other unique identification.
X-STREAM 2 series gas analyzers are intended to be used as analyzers for industrial purposes. X-STREAM is the process analyzer protected by the flame proof enclosure. Internal volume of the analyzer can be purified by the synthetic gas or by the inert gas through flame arrestors installed on the inlet and the outlet of purification path. The analyzer is equipped by the breathing devices formed by the flame arrestor in the case of gas analyzed pressure above from 1100 hPa to 1500 hPa. Analyzer marked as “….X2FD” is equipped with the alpha-numeric display. Analyzer marked as “…XEFD” is equipped with the graphic display. Other small modifications concern the internal mechanical layout as well as the internal interconnection of I/O boards not having influence on the level of protection.
Instruction for use - see documents HASXMDE-IM-EX and HASXEDE-IM-EX.
Special conditions for safe use, including dimension of the flameproof gaps, are specified at the chapter 1.5.1 of the Instruction Manual of the manufacturer No. HASXMDE-IM-EX (4/2009) and HASXEDE-IM-EX (9/2009).