IECEx BAS 09.0118Ex ia d IIC T4 Gb (-20 °C ≤ Ta ≤ +55 °C )orEx ia d IIC T3 Gb (-20 °C ≤ Ta ≤ +55 °C )
This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate, was assessed and found to comply with the IECEx Quality system requirements. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
This Unit verification certificate is issued as verification that the Apparatus identified on page 1, was assessed and tested and found to comply with the IEC Standard list below. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
The apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards:
As this is a Unit Verification Certificate, no QAR is applicable as this certificate is specific to the items listed by serial number or other unique identification.
NOTE: This certificate was CANCELLED on 11th January 2017 at the request of the issuing IECEx Certification Body (ExCB) following their consultation with the Applicant. This cancellation does NOT affect products manufactured or installed prior to this date. Any comment may be directed to the Applicant or ExCB.
The G-TECTA 4GP Gas Detector is designed to monitor the concentration of oxygen (deficiency), toxic or flammable gas, and provide visual, audible and physical (vibrator) warnings if preset limits are exceeded.The presence of any sensor is optional and the maximum possible is one flammable, one oxygen and two toxic.It comprises electronic circuits on printed circuit boards, a display, various LED indicators, a pump, a vibrator, a sounder and a rechargeable lithium-ion battery, all contained in a plastic enclosure providing a degree of protection of at least IP20.The flammable sensor used is either type VQ500 series by E2V Technologies, to Certificate IECEx SIR04.0014U, or 4P series by City Technology to Certificate IECEx SIR04.0013U, with a Code of Ex d IIC Gb and a maximum permitted ambient temperature of +55ºC. These sensors are certified to IEC60079-0:2007 and IEC60079-1: 2007.The IR Sensor used is type MSH-P Series by Dynament to Certificate IECEx SIR05.0053U, with a Code of Ex d IIC Gb and a maximum permitted ambient temperature of +60ºC. This sensor is certified to IEC60079-0:2004, IEC60079-1:2003 and IEC60079-26:2006.Marking of the Gas Detector is in accordance with IEC 60079-0:2007 and is:Ex ia d IIC T4 Gb (-20ºC ≤ Ta ≤ +55ºC).The apparatus is not designed for use in oxygen enriched atmospheres.Charging conditions:The battery must only be recharged in a non-hazardous area, using a Linde charger with an output (Um) of 9V.Non-rechargeable instrument:There is a non-rechargeable version of the instrument which has Temperature Class T3 and is therefore marked:Ex ia d IIC T3 Gb (-20ºC ≤ Ta ≤ +55ºC).The batteries must only be changed in a non-hazardous area.