This certificate is issued as verification that a sample(s), representative of production, was assessed and tested and found to comply with the IEC Standard list below and that the manufacturer's quality system, relating to the Ex products covered by this certificate, was assessed and found to comply with the IECEx Quality system requirements. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
This Unit verification certificate is issued as verification that the Apparatus identified on page 1, was assessed and tested and found to comply with the IEC Standard list below. This certificate is granted subject to the conditions as set out in IECEx Scheme Rules, IECEx 02 and Operational Documents as amended.
The apparatus and any acceptable variations to it specified in the schedule of this certificate and the identified documents, was found to comply with the following standards:
As this is a Unit Verification Certificate, no QAR is applicable as this certificate is specific to the items listed by serial number or other unique identification.
Handheld, battery powered instrument using photo-ionisation technology for the detection of Volatile Organic Compounds (VOC's) which can be dangerous from both a poisoning and explosive perspective. The TIGER uses a Photo-Ionization Detector (PID) to measure gas concentrations. and a patented fence electrode technology to minimise the effects of moisture and contamination, avoiding the need for compensation. The TIGER can be connected directly to a PC via a standard USB cable facilitating data download. An audio output, vibration and LED's are used to indicate alarm conditions. Orange and Red LED's indicate High and Low conditions respectively. The Instrument comprises three printed circuit boards, a pump motor, a vibration motor, certified PID Sensor and a battery pack all housed inside a plastic enclosure. There are two battery pack options; namely a rechargeable lithium ion battery pack using a SAFT MP174565 Integration cell and a non-rechargeable pack using three alkaline manganese cells, type Duracell Procell MN1500 which are user replaceable.
Input Parameters: USB port: Um = 6V. Charging connector for Li-Ion Battery Pack: Um = 6V.
Refer to attached schedule of drawings.
The PhoCheck Tiger must be functionally checked prior to entering a hazardous area after every occasion when a connection has been made to the USB port. The instrument must complete its start up routine and display legible readings. If the LCD display fails to show an intelligible and uncorrupted display the instrument must not enter a hazardous area.
Do not make any connection to the charger port or USB port of this instrument in a hazardous area.